Question about GE Medical
Make sure your print/record speed is 25mm/sec. Most machines have settings of 50mm/sec or 25mm/sec. I bet yours is set to 50mm/sec. RJ
Posted on Sep 28, 2017
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
SOURCE: EKG Machine
3 lead machines are designed to monitor the cardiac rhythm cycle, in 3 of the 12 primary leads. An attempt to use this limited information as a true diagnostic tool would be incomplete and perhaps misleading.
A full 12-lead machine is diagnostic, and looks at the elecrical information from all sides ("walls", ie anterior, septal, inferior, lateral, posterior) of the heart and gives detailed information regarding ischemia and or infarction of the cardiac muscle, electrical conduction system function, electrical axis, chamber enlargement, drug effects, and more.
Hope this helps.
Posted on May 08, 2009
You should purchase a printer kit. This is a common problem I have encoutntered on this machine and this machine is identical to the Seca CT6I. I have ordered the printer kit from Seca in the UK, the part number is 9740490024 at a cost of £70. It is a straightforward repair.
Posted on Sep 11, 2009
The heat on the print head can fade over time giving you a poor print. also the feed roller can wear and give poor pressure to the print head. thw way to diagnose it is to put a peice of electrical tape on the print roller and print an ekg. if it prints well in the area where the tape was, it is the print roller. if there is no change it is the print head.
email@example.com. www.overstockme.com. Jeff Watson
Posted on Jan 01, 2010
SOURCE: Our MAC 1200 ekg machine
1.Press the setup Button
2.Select operating mode and press enter.
3. Repatedly perss the down arrow until it reaches to the "START AT QUEUE MARK" and perss enter. Then select NO and press enter ( if it is selected Yes then before each recording the recorder advance the paper to the beginning of the next page)
4. Press the Start button to exit the setup.
From the next EKG it will print from the first page.
Posted on Aug 11, 2011
Tips for a great answer:
Dec 21, 2015 | Medical
Sep 27, 2011 | Ge Medical Systems Ge Marquette Mac 1200...
Aug 09, 2011 | Ge Medical Systems Ge Marquette Mac 1200...
Apr 26, 2011 | GE Medical
Dec 07, 2010 | Ge Medical Systems Ge Marquette Mac 1200...
Sep 01, 2010 | Ge Medical Systems Ge Marquette Mac 1200...
Apr 19, 2010 | Ge Medical Systems Ge Marquette Mac 1200...
Nov 17, 2009 | Ge Medical Systems Ge Marquette Mac 1200...
Apr 05, 2009 | Schiller Welch Allyn Cardiovit At 2...
Mar 19, 2009 | Ge Medical Systems Ge Marquette Mac 1200...
1,657 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: