Question about Tasco 750 Monocular Microscope
I have a infinity light microscope and when I raise the condenser (or adjust the focus) to a certain height many spots with a bluish/red tinge appear in the field of view. I have turned the eyepieces, entire condenser (with attached filter), objectives and the spots remain stationary. I can't be sure whether it is dust or not, and I'm concerned they may be scratches, although the microscope is brand new. What do scratches/lens imperfections look like under the microscope? Also is there any other way of locating the problem?
There is five part in microscope
1. Eye piece
2. Binocular tube
4. magnification changer or joom accembaley
5. illumination segment
Try to make it out which part is gives a effect to your vision.
as per my suggesition illumination asembely you just clean it first.
keep in touch
Posted on Feb 20, 2008
Hey there to you too!
I suggest you don't have a problem at all. You are focusing on a ground glass filter somewhere in the light train. This serves the purpose of diffusing the light to give even illumination across the field of view.
When setting up, firstly use low power and a simple, easy specimen. Focus on this then rack up the condenser untill you see your annoying speckles. Rack up the condenser a little more untill the speckles just go out of focus. This is the position where your condenser should be for normal use.
Setting up the light source is the most important pre requasit for a good image, and there is a lot more to it than what I have already mentioned.
If I can be of any further help......Don't hesitate
Cheers and good hunting...Dave
Posted on Mar 10, 2008
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
Dec 28, 2014 | Science & Laboratory
Nov 18, 2014 | Science & Laboratory
May 16, 2011 | Computers & Internet
Jul 28, 2010 | Leica 13596010 Microscope
Dec 27, 2009 | Science & Laboratory
Aug 30, 2005 | Nikon Coolpix 4500 Digital Camera
237 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: